This automated optical fiber tensile testing system is made for operator skill impartial use in production and lab configurations. Two fundamental model configurations enable the user to pick between a set up for optical fiber ribbon machine along with additional mandrels for rotary high strength testing.
The linear edition is made for production evidence testing with forces up to 22 N. Motorized clamps with carefully designed rubberized inserts be sure that the fiber coating will not be damaged inside the screening process while concurrently getting rid of the necessity for an external air supply. Levels of tugging force, tugging velocity and hold time are automated and conveniently offered to the operator with the built-in LCD touch-screen control. Data from tests can be stored and accessed via the USB dock or Ethernet user interface.
Mandrels of the identical design like NYFORS? optical fiber recoaters can be easily attached to the proof tester within the rotary higher power settings. The fiber is covered around the two mandrels which assistance it having a unique clamping system. By doing this higher loads can be used to the fiber, allowing breaking power tests below controlled conditions.
Adjustable clamp placement facilitates testing using the highest force levels as well as testing of short fiber lengths. The lightweight and light weight style makes the instrument easy to move around in a laboratory environment and to incorporate within a production work bench where space is at a premium. The machine is a compact and light-weight tool ideal for testing the potency of optical fibers up to 30 N. The machine is developed to fulfill all types of applications in which a strength check has to be carried out.
The machine is made with all the user in mind. The optical fiber ribbon machine is definitely placed in the guided linear clamps, and also the testing begins instantly having a One Touch go switch. The built in microprocessor controls all the important parameters such as pulling force, pulling speed and hold time. This permits to get a totally controlled test each and every time.
The machine can be linked to another Computer, which allows access to all programmable parameters and settings. The machine is tailored to stand-alone use or even to the incorporation into different subsystems. The machine comes with an substantial ability of signing and conserving information such as tugging force, tugging speed and fiber diameter along with other guidelines.
Two recoat injector designs can be purchased. The PTR307 uses a computerized water pump to inject the recoat materials. The quantity of materials dispensed by the automated injector is managed by hand through the top-mounted “Inject” switch or programmed to the machine utilizing the tablet pc control. The PTR307B features a manual recoat shot system that requires the consumer to manually dispense the recoat materials into the mold cavity; a replacement injector for the PTR307B recoater is accessible below. Make sure you note that this automatic injector is just suitable for high-index recoat material, while the TCC laser printer for cable works with both reduced- and- index bpabim materials; both are marketed individually below. Make sure you contact Tech Assistance for additional information.
he manual injector can be installed to compatible fiber recoaters using the 4-40 anchoring screws on the recoater real estate (see picture for the right). Make use of a 3/32″ hex key to secure the injector before use. For connecting the PTRRRM to the recoater mold, tighten up the connector after the eco-friendly plastic tubes, then loosen by a 1/4 turn to enable rotation.
The injector comes with a syndication valve as well as 2-position selection handle for directing the flow of recoat materials. A knurled dispensing screw with an inner plunger behaves as a syringe for the recoat materials. To fill the syringe, point the lever downward (i.e., towards the recoat bottle), then spin the knurled dispensing attach counterclockwise till it spins easily to fill up the syringe (shown in the photo towards the right). Then, to inject the recoat material to the mold, point the lever horizontally (i.e., facing the knurled attach) and rotate the attach clockwise until nearby the finish of the journey range is achieved. Steer clear of bottoming the dispenser as this may harm the interior plunger; also take care when re-interesting the threads to prevent go across threading the dispensing screw. Several fill/inject actions may be required until air is displaced within the system. Use camera lens tissue as well as an acetone or alcoholic beverages cleaning solution to gather any extra recoat material that runs from your mold.